Looking to add some serious digital inputs to your VINT Hub? This module has four active-high digital inputs ideal for reading switches, buttons, digital sensors, and other devices with a logic-level output. Each input can withstand up to 30V from the devices it's connected to, and they trigger faster than ordinary VINT digital inputs, making it more likely to trigger from very brief changes.
The VINT port on this board is isolated from the digital inputs, improving stability by removing unintentional ground loops from the system. Each input is also isolated from one another, reducing the amount of cross-channel interference.
If you need a module with more isolated digital inputs per VINT port, or if you want a simpler, more cost effective solution, have a look at the Comaptible Products tab.
|Current Consumption Min||28 μA|
|Current Consumption Max||2.8 mA|
|Number of Digital Inputs||4|
|Low Voltage Max (False)||1.3 V DC|
|High Voltage Min (True)||2.5 V DC|
|Low Voltage Trigger Length Min||0.2 ms|
|High Voltage Trigger Length Min||0.05 ms|
|Digital Input Current Max||* 4.5 mA|
|Digital Input Voltage Max||30 V DC|
|Recommended Wire Size||16 - 26 AWG|
|Operating Temperature Min||-40 °C|
|Operating Temperature Max||85 °C|
*Note: This value varies with input voltage. See the technical section of the User Guide for more information.
|Digital Input Isolated||DigitalInput||0 - 3|
|DigitalInput||Visual Basic .NET||Windows||Download|
|Date||Board Revision||Device Version||Comment|
|June 2017||0||102||Product Release|
Welcome to the DAQ1300 user guide! In order to get started, make sure you have the following hardware on hand:
Next, you will need to connect the pieces:
Now that you have everything together, let's start using the DAQ1300!
In order to demonstrate the functionality of the DAQ1300, the Phidget Control Panel running on a Windows machine will be used.
The Phidget Control Panel is available for use on both macOS and Windows machines. If you would like to follow along, first take a look at the getting started guide for your operating system:
Linux users can follow the getting started with Linux guide and continue reading here for more information about the DAQ1300.
After plugging the DAQ1300 into your computer and opening the Phidget Control Panel, you will see something like this:
The Phidget Control Panel will list all connected Phidgets and associated objects, as well as the following information:
The Phidget Control Panel can also be used to test your device. Double-clicking on an object will open an example.
Double-click on a Digital Input object in order to run the example:
General information about the selected object will be displayed at the top of the window. You can also experiment with the following functionality:
The DAQ1300 features four isolated digital input terminals. Each input is isolated from all other inputs, which allows a single device to connect to a variety of sources, and prevents cross-channel interference.
Each input is active-high, meaning the (+) terminal must be pulled at least 2.5 volts above the (-) terminal to register as TRUE.
Here are a few things you can do with the DAQ1300 Digital Input Phidget
To interface a switch with the DAQ1300, an external voltage source must be provided to pull the input HIGH.
All inputs on the DAQ1300 are isolated from one another, so a variety of signals from different sources can be monitored with a single device.
The DAQ1300 can be used without additional circuitry to detect the presence of a voltage on a line (from a power source or a signal), by wiring the input terminals to the line and its corresponding ground.
The current consumption of this device varies depending on the voltage applied at the input. It ranges from 1.6mA to 4.5mA, according to the curve below:
This Phidget is a smart device that must be controlled by a VINT Hub. For more information about VINT, have a look at the VINT Primer. You can use a Phidget Cable to simply and easily connect the two devices. Here's a list of all of the different VINT Hubs currently available:
Use a Phidget cable to connect this device to the hub. You can solder multiple cables together in order to make even longer Phidget cables, but you should be aware of the effects of having long wires in your system.
|Image||Part Number||Price||Number of Digital Inputs||Low Voltage Max (False)||Low Voltage Max (True)||High Voltage Min (False)||High Voltage Min (True)|
|1010_0||$80.00||8||—||900 mV DC||4.2 V DC||—|
|1011_0||$50.00||2||—||800 mV DC||2.1 V DC||—|
|1012_2B||$95.00||16||900 mV DC||—||—||4.2 V DC|
|1018_2B||$80.00||8||—||900 mV DC||4.2 V DC||—|
|1019_1B||$110.00||8||—||900 mV DC||4.2 V DC||—|
|1065_0B||$75.00||2||—||800 mV DC||2.1 V DC||—|
|1073_0||$140.00||8||—||900 mV DC||4.2 V DC||—|
|1203_2B||$95.00||8||—||900 mV DC||4.2 V DC||—|
|DAQ1200_0||$12.00||4||—||1.5 V DC||3.5 V DC||—|
|DAQ1300_0||$20.00||4||1.3 V DC||—||—||2.5 V DC|
|DAQ1301_0||$50.00||16||1.3 V DC||—||—||2.5 V DC|
|HUB0000_0||$30.00||6 (Shared)||—||1 V DC||1.8 V DC||—|