I noticed recently that all 4 channels of my Phidget 1048 has poor 0.2deg C resolution. When temps fall slowly, I can see this with Phidgets Control Panel. Basically temps hold than then step downward 0.2C. Weird thing is, the temp does change slightly as it samples. I just updated the control panel tonight (build March 4th, 2019). The previous build was March 7th, 2018 and had the same issue. I tried a different USB on my laptop (didn't help). I'm running Windows 7 Pro 64bit.
Any clues as what would cause this? Anyone else have an issue like this? Could my 1048 gone bad?