1127 User Guide
|Go to this device's product page|
Checking the Contents
You should have received:
In order to test your new Phidget you will also need:
Connecting the Pieces
Testing Using Windows 2000 / XP / Vista / 7
Make sure you have the current version of the Phidget library installed on your PC. If you don't, follow these steps:
- Go to the Quick Downloads section on the Windows page
- Download and run the Phidget21 Installer (32-bit, or 64-bit, depending on your system)
- You should see the icon on the right hand corner of the Task Bar.
Running Phidgets Sample Program
Double clicking on the icon loads the Phidget Control Panel; we will use this program to ensure that your new Phidget works properly.
The source code for the InterfaceKit-full sample program can be found in the quick downloads section on the C# Language Page. If you'd like to see examples in other languages, you can visit our Languages page.
Updating Device Firmware
If an entry in this list is red, it means the firmware for that device is out of date. Double click on the entry to be given the option of updating the firmware. If you choose not to update the firmware, you can still run the example for that device after refusing.
Double Click on the icon to activate the Phidget Control Panel and make sure that the Phidget InterfaceKit 8/8/8 is properly attached to your PC.
Testing Using Mac OS X
- Go to the Quick Downloads section on the Mac OS X page
- Download and run the Phidget OS X Installer
- Click on System Preferences >> Phidgets (under Other) to activate the Preference Pane
- Make sure that the is properly attached.
- Double Click on in the Phidget Preference Pane to bring up the Sample program. This program will function in a similar way as the Windows version.
For a step-by-step guide on getting Phidgets running on Linux, check the Linux page.
Using Windows Mobile / CE 5.0 / CE 6.0
For a step-by-step guide on getting Phidgets running on Windows CE, check the Windows CE page.
The human eye is less sensitive to changes in light intensity than the 1127, but is able to see a wider range. The Human eye range is from 50 µlux (starlight) to 100 klx (extremely bright sunny day). The 1127, on the other hand, is able to measure from 1 lux (Moonlight) to 1000 lux (TV studio lighting) The 1127 is able to detect higher frequency fluctuations in light levels than the human eye. If you notice noise on the signal that you cannot perceive yourself, it is probably due to incandescent light flicker, or other varying light sources. This sensor is designed to respond to visible light, and it can sense light from concentrated sources like laser pointers (although be careful with high-power lasers, as they could damage the sensor). It will also have a very muted response to IR light that is close to the visible spectrum (700-800 nm). The 1127 is Non-Ratiometric which means that you cannot rely on the sensor saturating at 1000 SensorValue. To be conservative, interpret SensorValue over 950 as saturated, with the true light level being unknown.
The 1127 uses the AMS104 light sensor package. The following graph illustrates the sensor's sensitivity to specific wavelengths of light.
The formula to translate SensorValue into luminosity is:
Other Interfacing Alternatives
If you want maximum accuracy, you can use the RawSensorValue property from the PhidgetInterfaceKit. To adjust a formula, substitute (SensorValue) with (RawSensorValue / 4.095) If the sensor is being interfaced to your own Analog to Digital Converter and not a Phidget device, our formulas can be modified by replacing (SensorValue) with (Vin * 200). It is important to consider the voltage reference and input voltage range of your ADC for full accuracy and range.
Each Analog Input uses a 3-pin, 0.100 inch pitch locking connector. Pictured here is a plug with the connections labelled. The connectors are commonly available - refer to the Analog Input Primer for manufacturer part numbers.
Phidget analog sensors do not have their own API- they simply output a voltage that is converted to a digital value and accessed through the "Sensor" properties and events on the PhidgetInterfaceKit API. It is not possible to programmatically identify which sensor is attached to the Analog Input. To an InterfaceKit, every sensor looks the same. Your application will need to apply formulas from this manual to the SensorValue (an integer that ranges from 0 to 1000) to convert it into the units of the quantity being measured. For example, this is how you would use a temperature sensor in a C# program:
// set up the interfacekit object InterfaceKit IFK = new InterfaceKit(); // link the new interfacekit object to the connected board IFK.open("localhost", 5001); // Get sensorvalue from analog input zero int sensorvalue = IFK.sensors.Value; // Convert sensorvalue into temperature in degrees Celsius double roomtemp = Math.Round(((sensorvalue * 0.22222) - 61.11), 1);
See the PhidgetInterfaceKit User Guide for more information on the API and a description of our architecture.
For more code samples, find your preferred language on the Languages page.
|Date||Board Revision||Device Version||Comment|
|May 2008||0||N/A||Product Release|
|April 2013||Product Discontinued.|